Cobb Tester Clamp System
Absorbency-testing device is used for determining the absorption of liquid mediums, by paper from the top and bottom sides within a predetermined time.
The IGT Cobb tester is an absorbency-testing device used to determine the absorption of liquids, such as water, aqueous solutions, oils, varnishes and others, by paper, paperboard and corrugated board within an accurately, predetermined time, in accordance with ISO 535, Scan P-37:77 Scan P12, Tappi 441 and Fefco nr. 7.
There are different types of Cobb testers available:
– for determinations on thin substrates like paper and light paper board. A special version is available for light structured papers (e.g. tissue), which can also be used for paper and board
– for determinations on substrates up to 15 mm thickness, like massive board or corrugated board
– for determinations on plaster board construction panels up to 22 mm thickness
– as the standard version with the additional option to measure the absorption curve until the liquid passed through the substrate.
The absorbency-ring is connected with a quick span system with a handle. The absorbency ring is lifted or pressed down by moving the handle.
The system can be adjusted for materials with different thickness by rotating the big nuts at top of the device.
Inner Ø 112.8 mm,
Max sample thickness 4 mm
Filling amount 100 ml
Test area 100 cm²
|Dimensions(L x W x H):||14 cm x 32 cm x 12 cm|
|Conformance with standards/W-leaflet:||—|
|Water bath and temperature:||—|
|Speed range and modes:||—|
|Interval time and number:||—|
|Printing shafts type and options:||—|
|Print length and width:||—|
|Sector type and options:||—|
|Data export options:||—|
|Accessories:||570.001 Quenching roller
570.002 Quenching plate
570.005 Blotting paper
570.007 Cup for 100 ml
|Inks or liquids:||Liquids|
|Compatible instruments:||570.000.800 EasyCobb
570.000.100 Cobb Tester SFT03t
570.000.322 Cobb Tester SFT03t for building boards up to 22 mm
|Measurement range:||Max sample thickness 4 mm|
|Light source/ power/angle:||—|
|Description of property:||—|
|Cleaning:||After using the device should be cleaned using a washing solution consisting of de-mineralized water and a washing-up liquid. After this, the device is to be rinsed with plenty of de-mineralized water. The liquid container is made of anodized aluminium.|
|Guarantee:||Standard one year|
|Space requirements:||0,09 m²|
|Standard delivery time:||4-6 weeks|